KEITHLEY 3390 MANUAL PDF
KiWAVE™ is a trademark of Keithley Instruments, Inc. All Keithley Instruments product names are .. Model Arbitrary Waveform Generator User’s Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 The exceptional signal quality of the Model is a result of its high resolution, . Refer to the Keithley Model Arbitrary Waveform Generator User’s Manual if it is not clear how to specify the waveforms. The figure below shows an.
|Published (Last):||17 June 2018|
|PDF File Size:||2.14 Mb|
|ePub File Size:||15.31 Mb|
|Price:||Free* [*Free Regsitration Required]|
Safety Precautions The following safety precautions should be observed before using this product and any associated instrumentation.
When installing equipment where access to the main power cord is restricted, such as rack mounting, a separate main input B-2 Error message definitions Section 1 Introduction In this section: Topic Page General information Introduction Handle adjustment You can adjust the carrying handle to suit Operation position Figure Operation handle position Position Section 2 Front Panel In this section: Topic Page Front panel description Front Panel Front panel Power key Press this key to turn The general purpose interface bus GPIB and Section 3 Rear Panel In this section: Topic Page Rear panel description Rear Panel Use this command to reset to a zero-phase reference Section 4 Setup Basics In this section: Topic Page Editing parameter values and settings Setup Basics Setting frequency or period Each waveform function has its Setup Basics Setting pulse high and low levels You can set Press the Utility key, Setup Basics Restoring factory default settings You can choose to reset Section 5 Waveform Output Operations In this section: Waveform Output Operations 2.
Refer to Section 4, Setup Basics for Setting up a square waveform To Wmin is the minimum allowable value determined Waveform Output Operations Figure Setting up a pulse waveform To Waveform Output Operations To generate an arbitrary waveform from a remote Waveform Output Operations enabled, the previously selected modulation is automatically disabled Press the soft key under Type; then Waveform Output Operations The default source is internal with a default Press the Graph key to view the Waveform Output Operations 5.
Press the soft keys corresponding with Waveform Output Operations Burst phase Burst phase defines the starting phase Setting repeat on or off for Section 6 System Operations In this section: System Operations To store a state from a remote interface: System Operations Performing self-test You can perform a self-test on the Section 7 Remote Programming In this section: Topic Page Remote interface operation Use the numeric keys to enter Remote Programming Front panel for a review of the front panel Standard Event register Bit Decimal value Appendix B Error Messages In this section: Error Messages Table B Command errors Error code Description Character Execution errors Error code Description Settings Execution errors Error code Description Data Query errors Error code Description Query Arbitrary waveform errors Error code Description Appendix C Application Programs In this section: Topic Page About application programs Application Programs To select linear sweep: C-4 Return to Section Index Numerics F 10 MHz connectors Service Form Model No.
Date Name and Telephone No.
Company Keeithley all control settings, describe problem and Specifications are subject to change without notice. All Keithley trademarks and trade names are the property of Keithley Instruments, Page of Go.
Artboard Artboard Artboard Artboard.